Acousto Elastic Sensor integrates MEMS technology.
October 2, 2013 -
RSS-X series is used to measure variations in compressive stress and load, allowing stress evaluation when material's modulus of elasticity is unknown or anisotropic and inhomogeneous. Compression of metal body, which consists of resin and protected housing arranged in concrete or iron braid of reinforcement, is measured with ultrasonic pulse-echo method. In case of concrete compressive stress sensor, ultrasonic sensor is included. Permanent temperature measurement is also integrated.
Additional information provided by manufacturer
|Original Press Release |
IBJ Technology Jäger
Colkwitzer Weg 7
Markkleeberg, , 04416
2D Rock Stress Sensor with MEMS
The RSS-X series of acousto elastic sensors from IBJ Technology is used to measure variations in compressive stress and load, which allows stress evaluation when the material’s modulus of elasticity is unknown or anisotropic and inhomogeneous. Typical examples are concretes and drill holes in the rocks, where these instruments can be used include:
• Tunnel supports
• Deformations of pillars
• Dams and nuclear power plants
• Structural members of buildings and bridges
• Piles and caissons
• Movements and landslides
The sensor detects the compression of a metal body. This metal body is 2 components resin and protected housing. The housing is arranged in concrete or in the iron braid of reinforcement.
The compression are measured with ultrasonic pulse-echo method. In the case of concrete compressive stress sensor, an ultrasonic sensor is included.
The RSS-X is offered in two models, RSS-X and RSS-2X, that differ in their measured axis.
RSS-2 X two systems owned by 90 ° turned. The horizontal and vertical stress is measured.
The sensors with Triaxial accelerometer with magnetometer (compass) or without can be equipped. Therefore, the alignment of the bore hole and the position of the sensor for crimping the expansion of cement can be determined.
The runtime in the measuring element is measured in high resolution with a separate stress meter. All ultrasound devices to the bolt measurement with a maximum frequency of 10 MHz is suitable for the measurement.
The term declines linearly with the compression in the elastic range.
Which the compressive stress sensor surrounding concrete is a compression of the measuring element. The change in the length of the measuring element is not only dependent on the load or stress, but also on the temperature.
Both factors taken into account the ultrasound measuring device for measuring the change in length.
A permanent temperature measurement is integrated into the pressure voltage sensor. This temperature sensor work as 1-wire sensor with ROM for the identification.
For universal use, the sensor for all devices for the measurement of bolt is suitable regardless of the manufacturer. The automatic temperature correction of these devices can be used, a second is incorporated independent temperature sensor.
The term due to the change in length and acousto elastical effect has reduced the influence of the surrounding concrete on the measuring element.