# Making High Resistance Measurements with DT8x Data Loggers

 Email | Comment Share

(Archive News Story - Products mentioned in this Archive News Story may or may not be available from the manufacturer.)

CAS DataLoggers
12628 Chillicothe Rd., Unit J
Chesterland, OH, 44026
USA

Press release date: August 24, 2011

For the DT8X Family of Data Loggers

CHESTERLAND OH - To measure the resistance of a particular device, the bestselling dataTaker DT8x family of data loggers uses the common technique of sourcing a highly accurate current through the device, measuring the resulting voltage, and then calculating the resistance as V/I. There are 2 current sources available at 2.5 mA and 200 uA, and to avoid a reduction in accuracy, the voltage measurement is performed without the high voltage attenuators so that the maximum voltage that can be measured is 3.0 volts, effectively limiting the maximum resistance that can be measured to approximately 10,000 ohms.

Sometimes it may be necessary to measure a larger resistance, for example in the case of a thermistor or potentiometer. The measurement range can be extended by wiring a known resistor in parallel with the resistance being measured. This will, however, reduce the resolution of low-resistance measurements.

In DeTransfer software, the command to make the measurement is as follows: 1R(4W,=1CV,W)

Then the actual resistance is calculated using the following expression: 2CV("R~ohm")=(Rp*1CV)/(Rp-1CV)

As shown above, users first read the combined resistance and store it in a channel variable (1CV), then calculate the value of R where Rp represents the value of the parallel resistor in ohms. As well as the 4-wire configuration shown here, a parallel resistor can also be used with a 3-wire or 2-wire resistance measurement. In all cases, the parallel resistor (Rp) should be located near the sensor®, as shown above, so that the lead resistances can be correctly compensated for.

If it is not practical to locate the resistor near the sensor, then it can be located at the logger end of the cable. In this configuration the best accuracy will be obtained by connecting the sense inputs (+ and -) across Rp (if its resistance is significantly less than R). If Rp is greater than R, then the sense inputs should instead be connected across R, although in this case the effect of cable resistance is likely to be negligible, given that both R and Rp are high resistances.

For further information on the dataTaker DT8x family of data loggers, other dataTaker data logging products, or to find the ideal solution for your application-specific needs, contact a CAS Data Logger Applications Analyst at (800) 956-4437 or visit the website at www.DataLoggerInc.com.

Contact Information:
CAS DataLoggers, Inc.
Chesterland, Ohio 44026
(440) 729-2570
(800) 956-4437
sales@dataloggerinc.com
http://www.dataloggerinc.com
 Email | Comment Share
 Contacts: View detailed contact information.

Name:
E-mail:
Comment title:
Comment:
To submit comment, enter the security code shown below and press 'Post Comment'.

 See related product stories Less ....
 See more product news in: Test and Measuring Instruments
| Featured Manufacturing Jobs
 Tools for you Print This Page Watch Company View Company Profile Company web site More news from this company E-mail this story to a friend Save Story
 Search for suppliers of Data Loggers & Data Logging Equipment Embedded Dataloggers Join the forum discussion at: Wired In

 Home  |  My ThomasNet News®  |  Industry Market Trends®  |  Submit Release  |  Advertise  |  Contact News  |  About Us Brought to you by Thomasnet.com        Browse ThomasNet Directory News provided by ThomasNet News® (TNN). TNN is a comprehensive source of new and timely product information in the industrial marketplace. TNN supplies new product information to the web sites, e-marketplaces and print publications that serve the industrial marketplace. For press release submissions please go to http://news.thomasnet.com/submitpr.html. BY ACCESSING, BROWSING AND/OR USING THIS WEB SITE AND/OR ANY WEB SITES PROVIDED BY ProductNews.com, YOU AGREE TO BE BOUND BY THE TERMS OF USE AGREEMENT.