Determining Thermocouple Accuracy on a dataTaker Data Logger

 Email | Comment Share
 Determining Thermocouple Accuracy on a dataTaker Data Logger Click Here to Enlarge Picture

(Archive News Story - Products mentioned in this Archive News Story may or may not be available from the manufacturer.)

CAS DataLoggers
12628 Chillicothe Rd., Unit J
Chesterland, OH, 44026
USA

Press release date: November 15, 2012

CAS DataLoggers Supplies Intelligent Universal Systems

CHESTERLAND OH - The spec sheets for any of the dataTaker DT8X series of loggers do not contain direct accuracy information on temperature measurements by thermocouple. Instead it offers a voltage accuracy based on the full scale of the voltage being measured. The Applications Specialists at CAS DataLoggers have assembled this brief tech article to show you how to use the voltage output to determine thermocouple accuracy using these flexible data logging systems.

A thermocouple uses 2 different metals to create a small electrical charge. The measurement of this charge can then be scaled to a temperature. The DT8X series of loggers is calibrated to ITS-90 standards when it comes to reading thermocouples. You can use this standard to determine the accuracy in degrees of the thermocouple.

Using the tables contained at: http://srdata.nist.gov/its90/menu/menu.html, the voltage output from a thermocouple can be determined for a certain temperature. This is done by selecting a temperature range for the correct type of thermocouple, then cross-referencing the resulting table.

For instance, a K-type thermocouple at 1200° Celsius will output 48.838 millivolts. If we check the accuracy of a DT8X logger on the spec sheet, we will see that the accuracy for DC voltage at normal operating temperatures (5°-40° C) is 0.1%, meaning that there may be a +- 0.048 mV variance. Looking at the table at 1200° Celsius, the difference between 1200 and 1201 is .037 mV, putting the accuracy of the dataTaker at that temperature range, just over +- 1°C.

The new Series 3 dataTaker monitoring solutions from CAS DataLoggers feature universal input channels for use with a wide range of input signal types to measure almost any signal including Temperature, Current, Voltage, 4-20mA Loops, Resistance, Strain Gauges, Bridges, Frequency, Digital, Serial and Calculated Measurements. These intelligent devices also offer powerful alarm and programming capabilities enabling them to process measurements and initiate actions on their own.

Boasting a rugged design and construction, dataTaker products are world-renowned for surviving rough handling and environments, built to last over years of constant operation. Their extensive communications arrays include RS232 with modem support, USB, Ethernet and USB memory stick ports so you can choose to connect to the device locally, remotely or over the Internet. This versatility is increased with the logger's multiple SDI-12 sensor networks, Modbus for SCADA systems, FTP and Web interface, so dataTakers are ready to be rolled out in any number of industrial machine monitoring projects.

For more information on the dataTaker DT8x series of Intelligent Universal Input Data Loggers, other dataloggers in the highly successful dataTaker line, or to find the ideal solution for your application-specific needs, contact a CAS Data Logger Applications Specialist at (800) 956-4437 or visit the website at www.DataLoggerInc.com.

Contact Information:
CAS DataLoggers, Inc.
Chesterland, Ohio 44026
(440) 729-2570
(800) 956-4437
sales@dataloggerinc.com
www.dataloggerinc.com
 Email | Comment Share
 Contacts: View detailed contact information.

Name:
E-mail:
Comment title:
Comment:
To submit comment, enter the security code shown below and press 'Post Comment'.

 See related product stories Less ....
 See more product news in: Test and Measuring Instruments
| Featured Manufacturing Jobs
 Tools for you Print This Page Watch Company View Company Profile Company web site More news from this company E-mail this story to a friend Save Story
 Search for suppliers of Data Loggers & Data Logging Equipment Join the forum discussion at: Wired In

 Home  |  My ThomasNet News®  |  Industry Market Trends®  |  Submit Release  |  Advertise  |  Contact News  |  About Us Brought to you by Thomasnet.com        Browse ThomasNet Directory News provided by ThomasNet News® (TNN). TNN is a comprehensive source of new and timely product information in the industrial marketplace. TNN supplies new product information to the web sites, e-marketplaces and print publications that serve the industrial marketplace. For press release submissions please go to http://news.thomasnet.com/submitpr.html. BY ACCESSING, BROWSING AND/OR USING THIS WEB SITE AND/OR ANY WEB SITES PROVIDED BY ProductNews.com, YOU AGREE TO BE BOUND BY THE TERMS OF USE AGREEMENT.